The characteristics of
Industrial temperature test, has been widely used in major oil and gas and other industrial temperature testing field;
Explosion-proof design: the flame-proof aluminum shell, intrinsically safe circuit board system;
Protection grade: IP67, fully sealed waterproof design;
Zigbee communication network to the coordinator or a router to its configuration and test;
Four and a half LCD display: temperature data, battery voltage;
LED indicator: reset instructions, directions configuration mode, access instructions, data acquisition;
Installation direction adjustable: by bringing union (or joint) installed on the protective casing, the adjustable direction.
Insert depth adjustable: by bringing the joint adjust temperature probe length, namely the insertion depth adjustment.
2. Performance indicators
21. The technical indexes
Working conditions: temperature, and 40 ℃ or less Tamb + 65 ℃ or less; The atmospheric pressure: 80 kpa to 110 kpa. Humidity, 5% RH to 95% RH
Power supply: 3.6 V19Ah batteries
Level of accuracy: 0.2 ℃ and 0.1 ℃
Sensor element: RTD Pt100
The output signal: Zigbee
Protection grade: IP67
Case material: cast aluminum
Protection casing material: 304 316 LSS
The process connection: union with protective casing
Range: - 200 ~ 600 ℃ range can be customized
Weight: 1 kg
Dimension: 390 * 116 * 137 (customizable)
F8913 module factory default name: well network ID: 123456, 58418: the default physical channel 13
XBEE module factory default network ID: 21 default communication channel: the default encryption enabled: 0
2. Explosion-proof authentication parameters
Explosion-proof number: CE18.2230 X
Explosion-proof marks: Ex ib Ⅱ C T4 Gb
Rated voltage: 3.6 V + / - 1% rated current: 50 ma rated power: 0.18 W
Directions for use
5.1. All transmitter through digital calibration, the different between the transmitter has the complete consistency and interchangeability.
5.2. In transit, all of the transmitter battery is in the off state, temperature transmitter installation on site, need to unscrew header back cover, battery access. Attention should be paid to tighten sheath to prevent leakage.
Install temperature transmitter (1) 5.3. The line pipe, pressure, discharge empty running. A blind than the sensor probe in the line pipe sleeve diameter is a little bigger (phi) 24 hole; (2) insert the tube sets of vertical from opening to half of the pipe diameter, and then along the hole week seal welding on the pipe; (3) blind pipe set inside into the heat conduction oil, about 2/5 of the volume of; (4) temperature sensor probe into blind pipe sleeve, adjust good header direction, with wrench will union tighten, pipeline transport.
5.4. Under the condition of temperature transmitter for a long time not to use, please open the header back cover, battery power.
6. Matters needing attention
6.1. This product use battery model have examined the explosion-proof certification, battery model for ER34615H; Replace the battery, it is prohibited to use other model battery.
6.2. Product antenna casing for the plastic material, potentially dangerous electrostatic charge! Use should be avoided in the friction! Please use wet cloth to wipe clean!
Contact Fax： 0552-4070672
Add： Sun Moon Science and Technology Park, No. 985, Xingzhong Road, High-tech Zone, Bengbu, Anhui, China
Force measurement and load cell
Wireless digital sensor
Temperature and pressure sensor
Angle, displacement, speed sensor
Oilfield measurement and control systemWeighing control system
Weighing logistics network solutions
Construction machinery sensor solutions
Oil and gas IoT solutions