5.1. All transmitter through digital calibration, the different between the transmitter has the complete consistency and interchangeability.
5.2. In transit, all of the transmitter battery is in the off state, temperature transmitter installation on site, need to unscrew header back cover, battery access. Attention should be paid to tighten sheath to prevent leakage.
Install temperature transmitter (1) 5.3. The line pipe, pressure, discharge empty running. A blind than the sensor probe in the line pipe sleeve diameter is a little bigger (phi) 24 hole; (2) insert the tube sets of vertical from opening to half of the pipe diameter, and then along the hole week seal welding on the pipe; (3) blind pipe set inside into the heat conduction oil, about 2/5 of the volume of; (4) temperature sensor probe into blind pipe sleeve, adjust good header direction, with wrench will union tighten, pipeline transport.
5.4. Under the condition of temperature transmitter for a long time not to use, please open the header back cover, battery power.
6.1. This product use battery model have examined the explosion-proof certification, battery model for ER34615H; Replace the battery, it is prohibited to use other model battery.
6.2. Product antenna casing for the plastic material, potentially dangerous electrostatic charge! Use should be avoided in the friction! Please use wet cloth to wipe clean!
Contact Fax： 0552-4070672
Add： Sun Moon Science and Technology Park, No. 985, Xingzhong Road, High-tech Zone, Bengbu, Anhui, China
Force measurement and load cell
Wireless digital sensor
Temperature and pressure sensor
Angle, displacement, speed sensor
Oilfield measurement and control systemWeighing control system
Weighing logistics network solutions
Construction machinery sensor solutions
Oil and gas IoT solutions
Copyright © 2020 Bengbu Riyue Instrument Research Institute Co., Ltd. 皖ICP备19017385号-1 Powered by: www.300.cn